MATEC Web Conf.
Volume 157, 2018Machine Modelling and Simulations 2017 (MMS 2017)
|Number of page(s)||7|
|Section||Modelling and simulation, structural optimization|
|Published online||14 March 2018|
Stimulation of nerve endings via medical device
Department of Machine and Mechanics, Faculty of Mechanical Engineering, University J. E. Purkyne, 40001 Usti nad Labem, Czech Republic
* Corresponding author: firstname.lastname@example.org
This solution describes solution of vibromechanism for stimulation nerve paths and nerve endings injured patients. Medical doctors stimulate nerve paths and nerve endings patients after injury of spine or after another injury where was broken nerve endings. These methods are in primary designed for tetraplegic and paraplegic patients specially for men. For this request was developed device and mechanism for vibrostimulation of nerve paths and nerve endings. In develop of construction has been paying attention to safety of mechanism. Next were taken into account the construction of other devices, especially their shortcomings in the operation on the accumulator. For mains powered vibrators 230 (110) Volt were assessed for safety and noise vibrating mechanism. The sum of all failures was proposed structure described below. The success rate of this vibration method is reported mostly between 60-80%.
Key words: vibration / excenter / amplitude / frequency / DC engine
© The Authors, published by EDP Sciences, 2018
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. (http://creativecommons.org/licenses/by/4.0/).
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.