MATEC Web Conf.
Volume 308, 20202019 8th International Conference on Transportation and Traffic Engineering (ICTTE 2019)
|Number of page(s)||5|
|Section||Transportation Infrastructure and Transportation Engineering|
|Published online||12 February 2020|
Minimum Required Distance of Strain Gauge from Specimen for Measuring Transmitted Signal in Split Hopkinson Pressure Bar Test
Mechanics of Materials and Design Laboratory, Department of Materials Engineering, Gangneung-Wonju National University, Gangneung, Gangwon-do 25457, Republic of Korea
a Corresponding author: firstname.lastname@example.org
The minimum required distance of the strain gauge on the transmitted bar of the split Hopkinson bar has been determined from the position of a metallic specimen via an explicit finite element analysis. The minimum required distance was determined when the strain-time profiles at r = 0, 0.5Ro and 1.0Ro, were coincident (r is the radial position and Ro is the radius of the bar.). The determined minimum required distance, f(x), is presented as a function of the relative specimen diameter to that of the bar (x = D/D0): j(x) = - 0.9385.x3 + 0.6624.x2 - 0.7459.x + 1.4478 (0.1 ≤ x ≤ 0.9). This result demonstrates the Saint-Venant's principle of rapid dissipation of localized stress in transient loading. The result will be useful for the design/modification of the pseudo-one-dimensional impact instruments that utilise a stress pulse transmitted through the specimen. The result will also allow one to avoid unnecessarily remote strain gage position from the specimen.
© The Authors, published by EDP Sciences, 2020
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.