MATEC Web Conf.
Volume 173, 20182018 International Conference on Smart Materials, Intelligent Manufacturing and Automation (SMIMA 2018)
|Number of page(s)||3|
|Section||Modeling, Analysis, and Simulation of Intelligent Manufacturing Processes|
|Published online||19 June 2018|
An effective method for quality control of the thermocouple
Shanghai Institute of Quality Inspection and Technical Research, Jiangyue Road, Minhang District, Shanghai, China
2 Shanghai Institute of Special Equipment Inspection and Technical Research, North NujiangRoad, Putuo District, Shanghai, China
3 Shanghai Institue of Measurement and Testing Technology, Zhangheng Road, New Pudong District, Shanghai, China
Corresponding author : firstname.lastname@example.org
As a basic testing element, the thermocouples are widely used in the safety and energy efficiency testing of the electronic and electrical products, new energy products and so on. The accuracy of all thermocouples can't be assured for they are mostly calibrated by sampling. In view of this situation, an efficient and convenient method for the intermediate check of the thermocouple is proposed in this paper. Taking 0 degree and 100 degree which can be easily obtained in the nature as the reference standard temperature, the deviation of the testing thermocouples and the reference standard temperature can be determined and the qualification rate of the thermocouples can be obtained. This method can promote the quality control of the laboratory.
© The Authors, published by EDP Sciences, 2018
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.