MATEC Web of Conferences
Volume 40, 20162015 International Conference on Mechanical Engineering and Electrical Systems (ICMES 2015)
|Number of page(s)||5|
|Section||Material properties and processing technology|
|Published online||29 January 2016|
Design and Fabrication of Low Cost Thick Film pH Sensor using Silver Chlorinated Reference Electrodes with Integrated Temperature Sensor
1 Research Center for Electronics and Telecommunication, Indonesian Institute of Sciences (LIPI), Bandung 40135, Indonesia
2 Department of Physics Education, Indonesian University of Education, Bandung 40154, Indonesia
This paper describes the design and fabrication of thick film pH sensor, in which the reference electrode has been formed by chlorination of Ag using FeCl3. The process was aimed to replace Ag/AgCl paste commonly used as reference electrodes. Fabricated using thick film screen printing technology on Al2O3 substrate, the pH sensor showed a measured sensitivity of -52.97, -53.17 and -53.68 mV/pH at 25°C, 45°C, and 65°C, respectively. The measured values were close to the theoretical Nernstian slope of -59 mV/pH 25°C.The sensor was also designed with an integrated Ruthenium based temperature sensor for future temperature compensation. The measured resistance temperature characteristics showed a linear reasponse over the range of 25 – 80°C. This miniaturised planar sensor should find wide application, especially in field water quality monitoring, replacing their glass type counterparts.
© Owned by the authors, published by EDP Sciences, 2016
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.