MATEC Web Conf.
Volume 252, 2019III International Conference of Computational Methods in Engineering Science (CMES’18)
|Number of page(s)||5|
|Section||Exploitation Machine Building|
|Published online||14 January 2019|
Technical aspects of SAW gas sensors application in environmental measurements
Lublin University of Technology, Faculty of Environmental Engineering, Nadbystrzycka 40B, 20-618 Lublin, Poland
* Corresponding author: email@example.com
The development of semiconductor technology and computer methods allows for the improvements in air pollutions measurement techniques. Besides the traditional techniques of air pollution measurement such as chromatography or dynamic olfactometry, the electronic nose (EN) gains increasing attention in this area. EN is a device which imitates the human olfactory sense. The main component of the device is an array of many nonspecific gas sensors. In this article, the most frequently used types of gas sensors in scientific studies were shortly characterised as: metal oxide semiconductors (MOS), conducting polymers, quartz microbalances (QCM) and surface acoustic wave (SAW) sensors, and others. Both their advantages and disadvantages were summarised. Technical aspects of measurements using SAW gas sensors arrays were presented, as well as measurement circuits were described.
© The Authors, published by EDP Sciences, 2019
This is an open access article distributed under the terms of the Creative Commons Attribution License 4.0 (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.