MATEC Web of Conferences
Volume 41, 20161st Mini Conference on Emerging Engineering Applications (MCEEA’15)
|Number of page(s)||4|
|Published online||01 February 2016|
A Theranostic Imaging prototype based on SiPM detectors for nanoparticles imaging during hyperthermia
Department of Biomedical Engineering, Technological Educational Institute of Athens, Athens, Greece
a Corresponding author: email@example.com
The combination of imaging and therapy has opened the very promising Theranostics domain with magnetic hyperthermia being a very promising domain. However, imaging systems should not interact with the magnetic field. In this work we have tested the recent C-series of SensL SiPM with 3mm pixel size, 4×4 arrays, coupled to different scintillators and irradiated with various gamma energies. The evaluation of the SiPM arrays shows that 1x1mm pixel size can be clearly resolved at PET energies for GAGG:Ce and CsI:Na and 1.5×1.5mm in SPECT imaging for CsI:Na. The best energy resolution was measured equal to 10.5% under 511keV irradiation for the 2×2mm GAGG:Ce; 16% under 511keV irradiation for the 1×1mm GAGG:Ce and 22% under 120keV irradiation for the 1×1mm CsI:Na. In addition, measurements with position sensitive photomultipliers have been carried out, to evaluate the effect of the magnetic field on the imaging performance of the system. While the effect of the magnetic field outside the coil is small, optimal images will be obtained if the imaging system is placed inside the coil, something that can be achieved only by using SiPMs.
© Owned by the authors, published by EDP Sciences, 2016
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.